Weighing tester is a kind of weighing equipment used in industry, agriculture, food and other industries today. It can help manufacturers produce qualified products more quickly. However, there are occasional problems in the process of use. Let's learn and solve with the staff of Jiawei Packaging.
When there is no weight display during the operation of the weight detector, you can check whether the relevant connector of the sensor is loose, deal with it in time, restart the device, and carry out the corresponding initial calibration. If the weighing value is unstable and there is a big jump, we can check whether there is debris on the weighing tray of the weight tester, or the detected residue is missing. If not, you can see whether the sensor is affected by other objects. influences. It should be noted that in order to ensure the stability of the weighing, we should regularly check the surrounding area of u200bu200bthe weighing tray and clean up the sundries above it in time.
In addition, when using the weighing machine, there are sometimes problems that the weight display is unstable but cannot be reset after starting up. This may be due to the influence of wind factors in the environment or the sundries on the roof. Lived the tray. And if the weight base on the display is large after power-on, it may be caused by the device being damp, and it can be restored after power-on for a period of time.
The above are some of the problems and solutions in the use of the weight tester. If you have more questions, please contact Jiawei Packaging Machinery Co., Ltd., and we will provide you with more solutions.Previous: What should I do if there is air in the packaging bag of the vacuum packaging machine Next: How to clean and maintain the weight checker?
Copyright © Guangdong Smartweigh Packaging Machinery Co., Ltd. | All Rights Reserved